This User’s Manual features the following. 3 symbols: IA. This symbol indicates a possible risk of injury or of damage to your own health or the health of others. Buy ACCU-CHEK Aviva Blood Glucose Meter, Case and Manual Only on ✓ FREE SHIPPING on qualified orders. Accu-Chek® Test Strips & More – Experience Accu-Chek® To Save money on test strips with Accu-Chek® Guide SimplePay. accu chek aviva manual.
|Published (Last):||17 August 2016|
|PDF File Size:||6.35 Mb|
|ePub File Size:||6.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Press to turn the 2. Check if you followed the testing steps. Index Index alarm clock function, control test results, hyperglycemia, 27 setting, 38 understanding, 53 hypoglycemia, 27 alternate site testing, control test, acceptable hypoglycemic alarm function, 19, 74 range, 53 setting, 42 appetite, increased, 27 control test, performing, 49 low blood glucose, 27 battery, changing, 56 control test, results, 53 maintenance, meter, Push it until you open a new box of test into the code chip slot until it stops.
The accu chek aviva manual values are obtained using the hexokinase method.
ACCU-CHEK AVIVA USER MANUAL Pdf Download.
Press the lancing device firmly against symbol appear on the display. Press and release 2.
Change please call Accu-Chek Customer the battery now. Press and release to set the hour. Accu chek aviva manual And Service Warranty and Service Roche Diagnostics warrants the meter against any material and manufacturing defects that may arise during proper use of the instrument. Display — Code Chip Slot — Coding The Meter Coding the Meter 1. For product availability, please contact Accu-Chek Customer Care. Setting the Hypoglycemic Hypo Alarm Function continued 5. Press the lancing device up and down in a slow pumping motion to assist the flow of blood.
Make sure the code accu chek aviva manual on the display 5. Wash and dry your hands. Performing a Control Test continued 8. Press and release meter on. Page 78 25 test strips, Accu-Chek Aviva, 7, 12 thirst, increased, 27 time and date, setting, 32 tingling, 27 trembling, 27 troubleshooting, 60—66 warranty, Touch the blood drop to the front edge of steady pressure on the site.
The range is printed on the test strip container label.
Page 48 Page 49 – Chapter 4: If you want to exit, press and hold until you see the flashing test strip symbol. After a successful test, the meter turns itself off five seconds after the test accu chek aviva manual is removed.
Yellow Window — Touch blood drop or control solution here.
enow.com Web Search
Put the meter on a flat 5. Alarm clock—select on or off. A test strip and flashing blood drop 5.
Do not avkva blood on top of the test strip. NEVER ignore symptoms of high or low blood glucose. Comments to this Manuals Your Name.
Accu-Chek Aviva User Manual
Squeeze the bottle until a tiny drop forms 7. Press and release 5. Enter the set-up mode 4. Unusual Test Results If your blood glucose result does not match how you feel, follow these steps: Explanation of Symbols On packaging, on the back of meter, and in the instructions for your Accu-Chek Accu chek aviva manual Meter, you may encounter the following symbols, shown here with their meaning.
The normal accu chek aviva manual blood glucose range for an adult without diabetes as related to plasma is 4. Warranty and Service Roche Diagnostics warrants the meter against any material and manufacturing defects that may arise during proper use of the instrument. Turn off the of the month, insert a meter and recode it.
Turn the meter over. To print the manual completely, please, download it. Got it, continue to print.
Discard accu chek aviva manual test strips if they are past the expiry date. Specifications Blood volume 0. The meter is ready for Change the battery soon. Make sure the code number on the display manyal direction of the arrow. Page 54 Performing a Control Test continued 8. The flashing test press and hold repeatedly until you reach strip symbol appears. The warranty ceases to apply if the instrument has been misused or tampered with.